For slower rates, you must use an external convert clock or an external convert clock timebase. daqdaqmxdata acquisitionnational instrumentsnicaiudllnidaqnidaqmxh Cancel Please login to add a comment or rating. I also see this error when generating LabVIEW code from the LabVIEW DAQ Assistant. An incorrect calibration might have been performed, or the calibration data in the EEPROM might have been corrupted. http://thetechvoice.com/error-code/common-x-error-codes-of-windows-xp.html
Thread To add a thread to your watch list, go to the thread page and click the "Add this thread to my watch list" link at the top of the page. KnowledgeBase; Request Support … DAQmx Base 1.0 on my PDA gives Error Code 15, what are some common solutions? … Contact NI © 2014 National Instruments Corporation. Error -200067Settling time constraints for the device could not be satisfied. Error -200220This "device identifier is invalid" error can occur if a USB task is aborted before DAQmx Base Stop Task.vi is called. http://digital.ni.com/public.nsf/allkb/368AE98647EEE86E86256EBE0067B167
Error -200237Hardware clocking error occurred. NIDAQmx error : The specified resource is reserved. > The operation could not be > completed as specified. Upgrade the installed version of NI-DAQ to a version compatible with the version used when saving the properties to take advantage of all the saved properties. Error -200150Channel is not available when the module is in parallel mode.
Error -200254Terminal for the device is invalid. Only the leading portion was saved. Error -200034Marker position exceeds the length of the subset. HW compatability is definitely not a problem.
Write samples more often. 4. Error 200088 Labview Error -200051Invalid identifier after a separator in the list entry. The syntax for a repeat statement is "
Any help will be much appreciated. Error -200192Switch device supports continuous scanning only. Error -200109Password is longer than four characters. Al "Jared Bancroft"
Refer to the documentation for proper syntax for connections involving channel ranges. Make sure your reference clock is connected and that it is within the jitter and voltage specifications. Labview Error Code 200088 Change the subset length to be longer than zero samples and a multiple of the alignment quantum. Error -200279 Daqmx Error -200234Buffer is too small for requested samples to be written.
Decrease the number of Pretrigger Samples per Channel, or increase Number of Samples per Channel. http://thetechvoice.com/error-code/comcast-set-top-box-error-codes.html If your data transfer method is interrupts, try using DMA or USB Bulk. 6. Error -200261An attempt has been made to use an analog trigger in multiple situations with differing properties. Solution: Note: The issue described in this document is resolved in the LabVIEW 7.1.1 Maintenance Release. Error Labview
For example, E Series devices use lines 0, 1, 2, and 4 on port 0 to communicate with the SCXI module. Error -200035Repeat loop is contained within too many levels of nested repeat loops. When I run the code I get error -200088.How can I … Why do I get Error -200088 When Running My NI-DAQmx Application in LabVIEW 7.1? http://thetechvoice.com/error-code/com-port-error-codes.html From: Sherryl Radbil Sherryl Radbil (view profile) 141 posts Date: 20 May, 2008 12:01:44 Message: 6 of 6 Reply to this message Add author to My Watch List View original format
I am using Matlab 184.108.40.206 R2006a, Data Acquisition Toolbox 2.8.1, Windows XP SP2, NI-DAQmx driver 7.5. If this is often the case, you may simply uninstall the program. Your application was unable to write samples to the background buffer fast enough for the device to get new samples at the specified sample rate.
Error -200032Subset length specified is not valid. For detailed information about cabling, refer to documentation. For example, DAQmx Base 3.1 for Windows is only compatible with NI-DAQmx 8.6 for Windows. Cannot configure these lines for output at this time.
Error -200232Property requested cannot be read. Error -200037"Clear trigger" instruction cannot be the last instruction of a "repeat" loop. Error -200128Property must have the same value for all repeated physical channels. http://thetechvoice.com/error-code/computer-error-codes.html In addition, this error may occur if NI-DAQmx was recently removed, or if NI-DAQmx Base was recently upgraded.
Alternatively, you can use multiple digitizers and SCXI modules in parallel mode. The specified gain and AI Minimum and/or AI Maximum would cause the device to exceed the hardware limit. Either the hardware between the source and destination terminals does not support the inversion, or other routes in the task might be interfering with this route. There are two likely causes for this error: 1.
Messages posted through the MATLAB Central Newsreader are seen by everyone using the newsgroups, regardless of how they access the newsgroups. Error -200125Lines 4 through 7 of this port are configured for output. For example, 20 MHz corresponds to a rate of 20,000,000 Hz. Error using ==> gui_fase_max('pushbutton1_Callback',gcbo,,guidata(gcbo)) A timeout occurred during GETDATA. ???
Get the Number of Possibly Invalid Samples property to see how many samples might be invalid. If you want to divide down an external clock, specifiy an external timebase source instead and set the clock source to be internal. Error -200007Record requested has not been acquired yet . Error -200004Input termination resistor for one or more channels was overloaded.
Tags are public and visible to everyone.